Copper strip corrosion tester for petroleum products
Copper Strip Corrosion Tester for Petroleum Products- Copper Strip Corrosion Tester
TR-TC-1020 Copper strip corrosion tester conforms to the ASTM D130 standard. the apparatus can determinate of the corrosiveness to copper of aviation gasoline, aviation turbine fuel, automotive gasoline, cleaners (Stoddard) solvent, kerosine, diesel fuel, distillate fuel oil, lubricating oil, and natural gasoline or other hydrocarbons having a vapor pressure no greater than 124 kPa (18 psi) at 37.8°C (930mmHg). It is a good laboratory special-purpose instrument in petroleum chemical industry, transportation, scientific research colleges and universities.
1.The instrument could be conducted four-way experiments, greatly improving the test efficiency.
2. The instrument provides a wide range of water bath, to make more stable results.
3.Digital display water temperature, temperature control accuracy of ± 1 ºC, temperature range from
room temperature to 100 ºC.
4. The instrument is simple with elegant structure.
1. Rated voltage: AC 220V±10% 50Hz
2. Power: 1500W
3. Precision of control: ±1ºC
4. Ambient temperature: 15~30ºC Humidity≤85%
Malfunction analysis and solutions
|Malfunction phenomenon||Cause analysis||Solutions|
|Panel lights off||1power switch damage|
2 The fuse wire becomes open circuit
|1. Replace power switch|
2.Check whether there is open circuit point in the electric circuit,replace the fuse wire after it is repaired
|Stirring motor broke||1.The motor is damaged|
2. Motor shaft stuck
3.Motor Power becomes open circuit
|1.Replace the motor|
2. Check jammed parts, motor to rotate the motor after the restoration of normal
3. Check the power cord, open motor to welding power supply cord
|Digital temp.controller shows abnormal temperature||1.temp.sensor is broken|
2.sensor opens weld
3.plug-in poor contact
|1.change the digital temperature controller|
2. Check the welded part, weld it again.
3. change plug-in
|Heating pipe stops working||1 Heating pipe becomes open circuit|
2 Heating pipe is damage
3Solid State Relay is damage
|1 Check whether the line is disconnected|
2 Measuring heating pipe resistance, change of heating tube
3 Replace the solid state relay
As a diversified manufacturer of high-tech products, it was founded in 1999, and devoted to the products, sales and technical service for various industries and fields like Energy analysis instrument, Oil, coal, steel, and geology; medical analysis and application instruments, biochemistry analyzer, elisa reader, blood analyzer,and urine; as well as laboratory application instruments,autoclave/sterilizer,freeze dryer, incubator,drying oven,balance,ICP,DRS,HPLC,gas chromatography,and spectrophotometer,meanwhile,TRYTE Technologies integrated and purchased some companies of PCB board, highway instruments at home and abroad. Our company has become the "one stop" instrument procurement service provider worthy of the name.
TRYTE technologies (TRYTE) mainly sell products by selling direct to customer, but also through distributors, resellers and manufacturer representatives, telephone sales and e-commerce sales, customers cover all over 80% of the world's countries and regions.TRYTE have solutions with rules adaptability according to the specific requirements of the design, from the instrument to some laboratories or departments.Whichever kind of solution you choose, you can fully trust it, because it is proved from 100, 00 successful results, leading development process, and decades of actual certification experience.
Why choose us?
1. We supply high quality products with reasonable price.
2. We are the professional manufacturer;
3. Advanced production machine;
4. Strict quality control system;
5. Ten people research&develop team;
6. We have professional sales team;